Spreader-CDR system for efficient, reproducible and frugal western blot
Cited 0 time in
- Spreader-CDR system for efficient, reproducible and frugal western blot
- Y R Cha; S Sajjad; K H Han; Tae-Sung Yoon; H S Shin; Sunghyun Kang
- Bibliographic Citation
- Electrophoresis, vol. 42, no. 14, pp. 1516-1520
- Publication Year
- Efficient antibody incubation is a vital step for successful western blot. During the incubation, a thin antibody-depleted layer is created around the blotting membrane, which limits antibody binding. Although the conventional batch shaking method is ineffective against it, this layer can be easily disrupted by cyclic draining and replenishing (CDR) of the antibody solution during membrane incubation. Previously, we introduced a closed and rotating cylindrical chamber as a tool to implement CDR for western blots (rCDR). A new open bucket-style chamber was devised for easier operation and the possibility of process automation. Instead of rotation as in rCDR, rocking it back and forth achieved the CDR antibody incubation (R-CDR). The chamber was then equipped with a spreader-rod to facilitate the uniform movement of the antibody solution across the membrane surface. Hence, it was named spreader CDR (S-CDR). Compared to the batch incubation method, both the S-CDR and R-CDR devices produced significantly enhanced signals and developed faster results. There were several additional benefits of using the spreader-rod, which included uniform antibody binding across the membrane, reduced usage of antibodies, and the ability to recover results even from mishandled, creased membranes. The S-CDR device ensures better blots and can be easily implemented in existing western blot protocols.
- Antibody incubation; CDR; S-CDR; Western blot
- Appears in Collections:
- 1. Journal Articles > Journal Articles
- Files in This Item:
Items in OpenAccess@KRIBB are protected by copyright, with all rights reserved, unless otherwise indicated.